Practical Use of Tinnitus Test Equipment Applied Tonal Bell Theory
Grant-in-Aid for Scientific Research (A)
Biomedical engineering/Biological material science
|Research Institution||Tokushima Bunri University|
FUJIMURA Tetsuya Tokushima Bunri University, Faculty of Engineering, Associate Professor, 工学部, 助教授 (80219047)
YOSHIDA Tomoji Tokushima Bunri University, Faculty of Engineering, Lecturer, 工学部, 講師 (80220656)
木村 章 (有)木村機械, 常務取締役
OHSAKI Katsuichiro Tokushima University, School of Medicine University Hospital, Professor, 医学部附属病院, 教授 (20116792)
KIMURA Akira Kimura Machinery Co., Ltd., Managing Director
|Project Fiscal Year
1995 – 1997
Completed(Fiscal Year 1997)
|Budget Amount *help
¥5,000,000 (Direct Cost : ¥5,000,000)
Fiscal Year 1997 : ¥1,100,000 (Direct Cost : ¥1,100,000)
Fiscal Year 1996 : ¥3,900,000 (Direct Cost : ¥3,900,000)
|Keywords||tinnitus test equipment / Tonal Bell Theory / subjective tinnitus / heptatonic scale / 耳鳴検査装置 / 音鐘理論 / 自覚的耳鳴 / 7音音階|
The objective of this research was to solve the both problems of using the conventional frequency pitch matching test and the continuous frequency pitch matching test by our proposed tinnitus test equipment based on Tonal Bell Theory, and develop a system that suits to an individual tinnitus patient by connecting with patients'information data base.
In order to utilize the on-line real time chinically, the device was made compact with the construction of an easy application exclusive electrical circuit. The target was to develop a system that can diagnose and construct patients'information data base connected into a computer.
As a result of this research, the test which considers the patients'treatment follow-up and testing tone handing were carried out easily on the computer screen. Thus recording during test and information management became very easy to carry out. The details are as follow ;
1. The developed system are composed by router, server network as well as tinnitus test equipment. Each equipment was connected through network.
2.The tinnitus test equipment was built-in into a noteboo type personal computer by design & construction of an exclusive electrical circuit.
Three test modes were established into a software for the test & diagnosis system :
(1) Conventional pitch matching test mode.
(2) A mode that can test automatically and its procedures decided beforehand with the reference on the past patients'information.
(3) A mode that allows freedom in the application of data base.
4.The results of the test recorded automatically by on-line real time, allowing the construction of patients'information data base.
Research Output (10results)